and can moves freely. • In electrodynamometer type wattmeter, moving coil works as pressure coil. • Hence moving coil is connected across the voltage and thus. Study of ammeter, voltmeter and wattmeter connections. • To investigate the errors .. an electrodynamometer for use as a wattmeter is shown in Fig. is kep. 5 Nov Electrodynamometer Wattmeter consists of two coils, pressure coil (PC) and current coil (CC). Pressure coil is connected across the circuit.
|Published (Last):||2 October 2015|
|PDF File Size:||4.34 Mb|
|ePub File Size:||13.28 Mb|
|Price:||Free* [*Free Regsitration Required]|
Used in Automotive and engineering Therefore we need to calculate the average of deflecting torque T d. The supply voltage applies to the moving coil. The fixed coil is divided into two equal parts and these are connected in series with the load, therefore the load current will flow through these coils.
Fixed coil is chosen as a current coil as it can be made wattmeyer massive. Let I 1 and I 2 be the instantaneous values of currents in pressure and electric current coils respectively.
Electrodynamometer Type Wattmeter | Electrical Study App by SARU TECH
The construction of a dynamometer instrument to some extent resembles the PMMC instrument. In a DC Circuit, the electrodynamoeter supplied to a load can be determined by measuring the load voltage and current and multiplying them together: The hydraulic radius and cross-sectional area of a channel is given by 4.
In modern wattmeter, CC is designed to only carry a current up to 20A. Instantaneous deflecting torque in electrodynamometer instrument is given as.
Notify me when new comments are added. So in electrodynamomete to limit the electric current we have connect the high value resistor in series with the moving coil.
Electrodynamometer Type Wattmeter
Therefore in circuit were power is quite high, Current Transformer shall be used for the measurement of power. Both moving coil and fixed coils are air cored.
Leave a Reply Cancel reply You must be logged in to post a comment. Controlling torque wattjeter provided by spring. Leave a Reply Cancel reply Your email address will not be published. Leave a Reply Cancel reply Your email address will not be published.
Consequently, current I v through the electrocynamometer coil is directly proportional to the load voltage. Basic construction of a dynamometer instrument. There are two types of coils present in the electrodynamometer. The pointer is fixed on the moving coil which is placed between the fixed coils.
Hence moving coil is connected across the voltage and thus the current flowing through this coil is always proportional to the voltage. Therefore for higher voltage Potential Transformer shall be used. Advantages of Electrodynamometer Type Wattmeter Following are the advantages of electrodynamometer type wattmeter and waftmeter are written as follows: The major difference from the PMMC construction is that the permanent magnet is replaced by two coils, as illustrated in figure 1 a.
Before we study the internal construction of electrodynamometer wattmeterit very essential to know the principle of working of electrodynamometer type wattmeter. They can be used for both to measure ac as well dc quantities as scale is calibrated for both.
As a high value of series resistance is connected in series with the PC, therefore PC circuit can be assumed purely resistive. Top Electrodynamometer Type Wattmeters Experts.
Dynamometer Type Wattmeter Working Principle | Electrodynamometer | Electrical A2Z
Gravity controlled system cannot be employed because there will be appreciable amount of errors. The scale of the wattmeter illustrated in Figure 2 is read directly only when the watgmeter switches are set to 1 A and V. There is uniform scale is used in these types of instrument as moving coil moves linearly flectrodynamometer a range of 40 degrees to 50 degrees on either sides.
Moving coil is connected across the supply voltage. The scale of the instrument can be calibrated to indicate watts, and thus it becomes a dynamometer wattmeter.
An electrodynamometer wattmeter have two types of coils. Now let the applied value of voltage across the electrodynamlmeter coil be Assuming the electrical resistance to wattmmeter pressure coil be very high hence we can neglect electrodynamoometer with respect to its resistance. The above expression gives the instantaneous deflecting torque of electrodynamometer wattmeter.
The deflection of the pointer of a dynamometer instrument is proportional to the current through the moving coil, but it is also proportional to the flux density of the magnetic field set up by the stationary coils. Related pages Measurement of Three Phase Power.
The pressure coil current in a low pf wattmeter may be as much as 10 times the value used for ordinary wattmeters. Then, when a current is passed through the moving coil, the moving coil and the meter pointer are deflected.